Is it possible to make an open loop crosstalk vs. frequency measurement using a 2700 series audio analyzer?
Yes. Drive channel A with the external sweep stimulus. On the Analog Analyzer panel, set the Function Reading Meter to Crosstalk with channel B selected, and set the BP/BR Flt Freq to Sweep Track. On the Sweep panel, set Source 1 to Anlr.Freq A, Data 1 to Alr.Level A, and Data 2 to Anlr.Crosstalk. See the accompanying test file XternalCrosstalkVsFreq.at27, which has all this set up for you.
Applies to: AP2700, APWin, ATS
Accompanies KB article "Open Loop Crosstalk vs. Frequency."